Dose mapping sensitivity to deformable registration uncertainties in fractionated radiotherapy – applied to prostate proton treatments
© Tilly et al.; licensee BioMed Central Ltd. 2013
Received: 27 September 2012
Accepted: 27 February 2013
Published: 14 June 2013
Calculation of accumulated dose in fractionated radiotherapy based on spatial mapping of the dose points generally requires deformable image registration (DIR). The accuracy of the accumulated dose thus depends heavily on the DIR quality. This motivates investigations of how the registration uncertainty influences dose planning objectives and treatment outcome predictions.
A framework was developed where the dose mapping can be associated with a variable known uncertainty to simulate the DIR uncertainties in a clinical workflow. The framework enabled us to study the dependence of dose planning metrics, and the predicted treatment outcome, on the DIR uncertainty. The additional planning margin needed to compensate for the dose mapping uncertainties can also be determined. We applied the simulation framework to a hypofractionated proton treatment of the prostate using two different scanning beam spot sizes to also study the dose mapping sensitivity to penumbra widths.
The planning parameter most sensitive to the DIR uncertainty was found to be the target D 95. We found that the registration mean absolute error needs to be ≤0.20 cm to obtain an uncertainty better than 3% of the calculated D 95 for intermediate sized penumbras. Use of larger margins in constructing PTV from CTV relaxed the registration uncertainty requirements to the cost of increased dose burdens to the surrounding organs at risk.
The DIR uncertainty requirements should be considered in an adaptive radiotherapy workflow since this uncertainty can have significant impact on the accumulated dose. The simulation framework enabled quantification of the accuracy requirement for DIR algorithms to provide satisfactory clinical accuracy in the accumulated dose.
KeywordsRadiotherapy Adaptive radiotherapy Dose tracking Dose mapping Dose accumulation Dose accumulation accuracy Deformable image registration Non-rigid image registration Protons
The patient geometry can change significantly between fractions during radiotherapy treatments. This motivates the use of image sets to monitor the treatment progress and to serve as basis for optional re-planning to better fulfil the treatment objectives.
The ideal monitoring scenario would be to score dose for each individual tissue part, or cell, throughout all delivered treatment fractions. The closest realisation of such a scenario is to calculate dose on images taken for each treatment occasion, and use deformable image registration (DIR) to map and accumulate the dose contributions onto a representative, single image dataset . Such dose distributions can be compared with the intended, planned dose distribution to provide a basis for corrective interventions in adaptive radiotherapy. The accuracy of the image registration will govern the accuracy of the mapped dose, and hence the relevance of the corrections. The validity of the accumulated dose, due to the image registration uncertainty, was recently debated by Schultheiss and Tomé . It is therefore of fundamental interest to determine the accuracy requirements of the image registration process, separate from other uncertainties, in particular with regards to the accuracy of the quantities derived from the finally mapped dose.
The input to the registration also contains uncertainties, e.g. image noise and uncertain segmentations. The registration uncertainty can be considered as uncorrelated to other treatment uncertainties such as organ motion effects or dose calculation uncertainties, and can thus be investigated separately. There exists a plethora of DIR algorithms in the literature, see e.g. [3, 4] for a review of available methods. Studies of the accuracy of different DIR algorithms applied to different body sites exist [5, 6] with reported average errors in the range of 1–5 mm. The accuracy of dose mapping has also been studied explicitly by several authors; Rosu et al. investigated dose grid resolution effects, Salguero et al. utilised the image registration inverse inconsistency and found a maximum dose mapping uncertainty of >30% of the prescription dose for a lung patient, Yan et al. related the lack of mass conservation in the registration to the dose uncertainty, Hub et al. estimated the dose uncertainty from the registration parameter uncertainty and Murphy et al. developed a method to sample image registration errors and demonstrated their effect on the mapped dose.
The aim of this study is to develop a methodology for determining the accuracy requirements for the use of deformable image registration as a basis for dose accumulation in fractionated radiotherapy. For this purpose we developed a simulation framework for dose accumulation where the uncertainty of the calculated treatment outcome (represented as dose, or as dose response determined with a biological models) as a function of the registration uncertainty could be investigated. The dosimetric impact of the registration uncertainty on the entire fractionated treatment has been studied by Risholm et al. who presented the uncertainty of the total physical dose for a single estimated registration uncertainty, while we study how the final dose and the related response would vary with the registration uncertainty.
We apply the framework to a hypofractionated treatment of the prostate with spot scanned protons where a “plan of the day” is assumed to be tailored to the CTV for each individual fraction. The clinical importance of dose accumulation for prostate treatments  was investigated in a recent publication by Wen et al. We hypothesise that the sensitivity to dose mapping uncertainty increases with steeper dose gradients and therefore we use dose distributions from two different spot sizes, one based on measurements at the local clinic and one with smaller spot sizes producing a sharper and less forgiving penumbra. We also investigate the accuracy requirement versus the required size of the planning margin used for construction of the planning target volume (PTV) from the clinical target volume (CTV).
The impact of the registration uncertainty on the accumulated dose, and its estimated response, can be studied by adding known uncertainties to the registrations used in mapping the deformations. The simulations mimic a clinical workflow where a “plan of the day” is tailored to the actual position of the CTV for the individual fractions. The total accumulated effect, i.e. the estimated response for the dose mapped from all fractions to a fixed reference geometry is scored for various degree of registration uncertainties. As we are specifically interested in the effects of the registration uncertainty we assume that the patient is imaged and setup without errors at each fraction. According to the van Herk margin scheme  this means that we simulate a situation yielding only the random contribution from DIR based dose mapping to the CTV to PTV margin. In a real clinical situation also other sources of uncertainty must be considered. In section A we describe the simulation framework and the modelling of the image registration uncertainty, and in section B we apply the uncertainty model to a hypofractionated prostate treatment with spot scanned protons.
A. Simulation of a fractionated treatment with dose mapping uncertainties
The dose distribution will vary from fraction to fraction because of organ motion and the “plan of the day” approach. We adopt a view where we with a given uncertainty can track the position of a tissue element at all treatments and accumulate the radiobiological effect fraction by fraction. The D F(r) then becomes a stochastic quantity.
The total equivalent dose, was used for analysis of the treatment DVHs. The tumour control probability, TCP, was calculated directly from S F(r), c.f. [17, 18]. The normal tissue complication probability, NTCP, was calculated using the Lyman model  with parameters from Burman et al. and Emami et al.. Also the equivalent uniform dose, EUD , was calculated from .
Modelling the image registration uncertainty
where is the true transformation, and the transformation error models the registration uncertainty. Neither or are in general known for a clinical situation and thus T (i) have to be treated as a stochastic quantity for which we assume that Registration algorithms apply various regularisation techniques to produce a well behaved and physically realistic transformation. Clearly, there is a local correlation of the error for points close together whereas for points further apart the error will be more uncorrelated. In modelling of this is mimicked by sampling uncorrelated errors at a sparse grid of control points which by means of a 3D cubic B-Spline interpolation is applied to the denser dose grid. The registration deviation amplitudes is sampled at the control points, independently for each coordinate according to a multinormal distribution with zero mean and standard deviation σ e. A low resolution of the control points will produce a slowly varying vector field and thus mimic a high degree of regularisation and vice versa for a high resolution of the control points.
B. Application of the simulation framework to a virtual prostate patient
The virtual patient model
The prostate is assumed to be incompressible and change location (but not shape) as the rectum and bladder change filling and shape. In prostate radiotherapy treatment it is common to also irradiate the lower parts of the seminal vesicles to the same dose as the prostate. The combined volume, prostate and lower parts of the vesicles, will constitute a rather convex volume which we for simplicity model as a sphere shaped CTV. We sample the displacement of the prostate centre for each fraction, according to a multinormal distribution with the standard deviations 0.4, 0.1 and 0.4 cm in the anterior-posterior, lateral and superior-inferior directions, respectively, consistent with the literature . The fixed geometry I F, into which the radiation effect is accumulated, is simply chosen to be the one with zero displacement of the prostate, cf. Figure 2a.
The dose grid resolution was set to 0.1 cm while two different grid sizes, 1.0 cm and 3.0 cm, was used for the control points in the registration error. Analysis of all T (i)(r), i.e. the patient model and the registration error, showed values of the Jacobian determinant, J(T (i)(r), r), in the interval [0.4, 2.0] for the lower resolution indicating an inverse consistent transform without folds or tears since J > 0. The T (i)(r) produced using the higher control point resolution included small regions with negative values of the J for e ≥ 0.5 cm indicating a transform that is not well regularised. It is desirable that the accumulated dose is independent of the choice of reference image I F and this requires that J > 0, which is not in general guaranteed by registration algorithms. The resulting dose mapping uncertainty was very similar for both resolutions but slightly more sensitive with respect to the image registration uncertainty for the 3.0 cm resolution. We have therefore concentrated on results for the 3.0 cm resolution.
Generation of treatment plans for simulation
The simulated treatments were assumed to be delivered with two opposed, scanned proton beams, indicated by the arrows in Figure 2a, aiming for a homogenous target dose.
The PTV was constructed by adding a margin m isotropically around the CTV.
Summary of the treatment planning objectives and the radiobiological parameters used in the evaluation
Minimise V(D<33.0 Gy)
Minimise V(D>34.7 Gy)
Minimise V(D>29.6 Gy)
Minimise V(D>29.6 Gy)
The “plan of the day” treatment scenario described above requires a tailored dose distribution for each target position for all patient instances, i.e. in equation (1). To save computation time and make the result less dependent of the intrinsic details of the optimiser we displace the dose distribution according to the fraction specific CTV displacement. This results in one optimisation per margin size and spot size regimen.
The scanned proton dose distribution was calculated using an in-house pencil beam algorithm with the depth dose and lateral scattering calculated according to Bortfeld  and Russell et al., respectively. Our normal beam model (NBM) parameters are based on data from Kimstrand et al.. Dose mapping in the presence of a sharp penumbra will be sensitive to the registration uncertainty, especially for targets where the loss of dose coverage can greatly affect the outcome. Dose distributions were therefore also generated according to a sharp penumbra beam model (SBM) whose in-air spot sizes are smaller and with zero divergence thus producing a sharper penumbra. The spot sizes of SBM correspond approximately to what modern commercial proton machines can deliver, e.g. see . The penumbra width is often reported as the distance that the dose falls from 80% to 20% of the target dose and here they were 1.16 and 0.88 cm for NBM and SBM respectively in the directions perpendicular to the beam axis. The spot size increase with treatment depth, due to multiple scatter of the protons, and note that a shallower target will have a sharper penumbra and thus more sensitive to dose mapping uncertainties.
The simulations were performed for ten different instances of the virtual patient where each instant had its own set of fraction specific CTV positions with corresponding deformation fields. A single treatment simulation, according to Figure 1, modelled the treatment of a patient instance with the prescribed 6.7 Gy × 5fx. To ensure adequate statistics, each treatment simulation for the same patient instance was repeated 40 times with different sampled image registration T (i) according to equation (5) for each value of the margin m (0.0, 0.1, …, 0.4 cm), image registration uncertainty 〈|T e|〉 (0.0, 0.05, …,0.4 cm) and beam models. This resulted in a total of 400 treatment simulations per combination of m, 〈|T e|〉 and beam model.
The DVH dependency on varying registration uncertainty
The spread of the DVHs for the OAR is less dependent of the planning margin and the penumbra width, although the overall dose level increases when any of these two parameters increases. The dose tend to be overestimated in the low dose region but underestimated in high dose regions which is clearly seen for the rectum and bladder in Figure 4. The reason for the underestimation for the high dose is similar as for the target dose, a dose mapping uncertainty is more likely to lower the dose (and vice versa for the low dose region).
To obtain the DIR accuracy such that |ΔD 95|95 is approximately 3% of the target dose, we found that the registration uncertainty 〈|T e|〉 must be less than 0.25 cm for NBM and SBM with SBM slightly more sensitive to 〈|T e|〉. If a planning margin of 0.1 cm is used, the 〈|T e|〉 registration uncertainty can be relaxed to 0.35 cm for NBM and 0.30 cm for SBM.
The value of |ΔV 80|95 was found to increase with increasing 〈|T e|〉 and weakly with increasing margin, as shown in the lower plots in Figure 6. The |ΔV 80|95 value was found to increase linearly as a function of 〈|T e|〉, and was approximately 1% and 4% of the ROI volume for 〈|T e|〉 = 0.1 cm and 0.35 cm, respectively. The |ΔV 80|95 increases weakly with increased margin and the difference between NBM and SBM is small. The results for the bladder |ΔV 80|95 were similar as the data found for the rectum.
The uncertainty in simulated radiation response
The sensitivity of the calculated radiobiological outcome quantities TCP and EUD for the CTV, and NTCP and EUD for the OARs, were determined for each treatment simulation. The chosen technique yields very low nominal NTCP values, less than 2% for the rectum and much smaller for the bladder. Accordingly, the EUD uncertainties for the risk organs were also very small, < 0.5 Gy.
The distribution of spatial dosimetric uncertainty with varying registration uncertainty
We have in this work developed a general framework for studying the influence image registration uncertainties have on the accuracy of the accumulated dose in fractionated radiotherapy. This framework was used to estimate the impact of the dose mapping uncertainties on the radiobiological quantities such as the TCP as well as the DVH quantities D 95 and V 80 for spot scanned proton therapy of the prostate. The results should be considered together to yield a complete view of the registration accuracy requirements. The relation between the TCP uncertainty and the DVH dose uncertainty can be illustrated by assuming a homogenous CTV dose where a 1% decrease would yield a 2.5% drop in the TCP. The decrease in D 95 is not the same as a decreased homogenous dose but it seems reasonable to require at least a 3% accuracy in D 95 and V 80. A |ΔD 95|95 of <3% would require that the mean absolute registration error should be better than approximately 0.25 and 0.20 cm for the larger and smaller spot size respectively, whereas a ∆TCP requirement of <2% would require a mean registration uncertainty better than 0.35 cm. An added planning margin of 0.1 cm significantly reduces the accuracy requirement to 0.35 cm and 0.30 cm for the same D 95 requirement for the larger and smaller spot size respectively. A |ΔD 95|95 <2% requirement yields a registration uncertainty requirement of 0.2 cm for the smaller spot size. A requirement of |ΔV 80|95 <3% would require a registration uncertainty of 0.25-0.30 cm regardless of spot size and margin. If the 3% requirement on D 95 is fulfilled for the intermediate penumbra, the summed volume with dose mapping uncertainty of more than 3%, in relative dose, V(3%), is small for the CTV but up to 15-20% for the organ at risks. The results indicate that the limiting accuracy requirement is that for the D 95. A relaxed penumbra, or an increased target margin, decreases the uncertainty in the CTV while it, due to the increased dose burden, increases the uncertainties in the organs at risk. Jaffray et al. highlighted the need for accurate dose accumulation for normal tissue and in that case the large V(3%) might be of concern, especially when accumulating dose to serial organs sensitive to local dose effects.
The result of our particular simulations would to some degree depend on the choices of treatment site, patient geometry and deformation method. A shallower target would sharpen the penumbra of the proton treatment and increase the registration accuracy requirement. We assume a registration algorithm without systematic errors and stochastic, uncorrelated random deviations with a spatial frequency resulting from regularisation such that the registration error could be represented as a B-spline vector field with normal distributed control points. We consider these conditions for the registration uncertainty to be rather general for any regularised algorithm with random errors, although its applicability for certain clinically used DIR routines must be checked explicitly. The dosimetric parameters are scored locally ROI by ROI making the conclusions less sensitive to the fact that the DIR uncertainty in reality is not evenly distributed in space. Studies show [5, 6, 34] that registration algorithms can in some cases, e.g. lung and liver CT-CT registration, have a mean absolute error close to 0.1 cm and thus fulfilling the 3% above stated requirement for D 95.
The uncertainty in the calculation of TCP obviously depends on the slope of the TCP sigmoid shape response curve, i.e. the γ 50 value. The parameters in this work were taken from the analysis of Cheung et al. where γ 50 = 2.2 (95% CI 1.1, 3.2). A higher γ 50 would increase the sensitivity to an image registration uncertainty and thus increase ∆TCP. Roughly, using γ 50 = 3.2 instead of 2.2 would raise ∆TCP from 3% to approximately 4.4%, assuming the equivalent relative drop in dose when starting from TCP = 80%.
Future research using the presented framework could include other deformation models and other delivery techniques such as rotational therapy with photons.
We developed a simulation framework that described the total accumulated dose and the related response as a function of the deformable image registration uncertainty.
When applied to a spot scanned prostate treatment, the accuracy of D 95 on the accumulated dose is a limiting requirement on the deformable image registration when performing dose mapping. A required accuracy of 3% in D 95 would require a mean absolute image registration error uncertainty of 0.20 cm. By increasing the CTV to PTV planning margin with 0.1 cm, the mean absolute error can be relaxed to 0.3 cm with respect to the D 95 requirement. A few algorithms have reported accuracy better than 0.2 cm uncertainty for CT-CT registrations and thus meet the above stated [5, 6, 34]. However, for more challenging registration problems, such as multi-modality registrations, more development might be required to improve the registration accuracy, or to establish its accuracy as to define the dose mapping contribution to margin requirements in adaptive radiotherapy.
The authors would like to thank Dr Alexandru Dasu for valuable discussions regarding radiobiological parameters in prostate radiotherapy.
- Yan D, Jaffray DA, Wong JW: A model to accumulate fractionated dose in a deforming organ. Int J Radiat Oncol Biol Phys. 1999, 44: 665-675. 10.1016/S0360-3016(99)00007-3.View ArticlePubMedGoogle Scholar
- Schultheiss TE, Tome WA, Orton CG: It is not appropriate to “deform” dose along with deformable image registration in adaptive radiotherapy. Med Phys. 2012, 39: 6531-6533. 10.1118/1.4722968.View ArticlePubMedGoogle Scholar
- Hill DL, Batchelor PG, Holden M, Hawkes DJ: Medical image registration. Phys Med Biol. 2001, 46: R1-R45. 10.1088/0031-9155/46/3/201.View ArticlePubMedGoogle Scholar
- Oliveira FP, Tavares JM: Medical image registration: a review. Comput Methods Biomech Biomed Engin. 2012, 1-21.Google Scholar
- Brock K: Results of a Multi-Institution Deformable Registration Accuracy Study (MIDRAS). Int J Radiat Oncol Biol Phys. 2009, 76: 583-596.View ArticlePubMedGoogle Scholar
- Kashani R, Hub M, Balter J, Kessler M, Dong L, Zhang L, Xing L, Xie Y, Hawkes D, Schnabel J, et al: Objective assessment of deformable image registration in radiotherapy: a multi-institution study. Med Phys. 2008, 35: 5944-5953. 10.1118/1.3013563.View ArticlePubMedPubMed CentralGoogle Scholar
- Rosu M, Chetty IJ, Balter JM, Kessler ML, McShan DL, Ten Haken RK: Dose reconstruction in deforming lung anatomy: dose grid size effects and clinical implications. Med Phys. 2005, 32: 2487-2495. 10.1118/1.1949749.View ArticlePubMedGoogle Scholar
- Salguero FJ, Saleh-Sayah NK, Yan C, Siebers JV: Estimation of three-dimensional intrinsic dosimetric uncertainties resulting from using deformable image registration for dose mapping. Med Phys. 2011, 38: 343-353. 10.1118/1.3528201.View ArticlePubMedGoogle Scholar
- Yan C, Hugo G, Salguero FJ, Saleh-Sayah N, Weiss E, Sleeman WC, Siebers JV: A method to evaluate dose errors introduced by dose mapping processes for mass conserving deformations. Med Phys. 2012, 39: 2119-2128. 10.1118/1.3684951.View ArticlePubMedPubMed CentralGoogle Scholar
- Hub M, Thieke C, Kessler ML, Karger CP: A stochastic approach to estimate the uncertainty of dose mapping caused by uncertainties in b-spline registration. Med Phys. 2012, 39: 2186-2192. 10.1118/1.3697524.View ArticlePubMedPubMed CentralGoogle Scholar
- Murphy MJ, Salguero FJ, Siebers JV, Staub D, Vaman C: A method to estimate the effect of deformable image registration uncertainties on daily dose mapping. Med Phys. 2012, 39: 573-580. 10.1118/1.3673772.View ArticlePubMedPubMed CentralGoogle Scholar
- Risholm P, Balter J, Wells WM: Estimation of delivered dose in radiotherapy: the influence of registration uncertainty. Med Image Comput Comput Assist Interv. 2011, 14: 548-555.PubMedPubMed CentralGoogle Scholar
- Wen N, Glide-Hurst C, Nurushev T, Xing L, Kim J, Zhong H, Liu D, Liu M, Burmeister J, Movsas B, Chetty IJ: Evaluation of the deformation and corresponding dosimetric implications in prostate cancer treatment. Phys Med Biol. 2012, 57: 5361-5379. 10.1088/0031-9155/57/17/5361.View ArticlePubMedGoogle Scholar
- Van Herk M, Remeijer P, Rasch C, Lebesque JV: The probability of correct target dosage: dose-population histograms for deriving treatment margins in radiotherapy. Int J Radiat Oncol Biol Phys. 2000, 47: 1121-1135. 10.1016/S0360-3016(00)00518-6.View ArticlePubMedGoogle Scholar
- Fowler JF: The linear-quadratic formula and progress in fractionated radiotherapy. Br J Radiol. 1989, 62: 679-694. 10.1259/0007-1285-62-740-679.View ArticlePubMedGoogle Scholar
- Bortfeld T, Jokivarsi K, Goitein M, Kung J, Jiang SB: Effects of intra-fraction motion on IMRT dose delivery: statistical analysis and simulation. Phys Med Biol. 2002, 47: 2203-2220. 10.1088/0031-9155/47/13/302.View ArticlePubMedGoogle Scholar
- Brahme A: Dosimetric precision requirements in radiation therapy. Acta Radiol Oncol. 1984, 23: 379-391. 10.3109/02841868409136037.View ArticlePubMedGoogle Scholar
- Nahum AE, Tait DM: Maximising control by customized dose prescription for pelvic tumours. Advanced Radiation Therapy: Tumour Response Monitoring and Treatmenr Planning. Edited by: Breit A. 1992, Heidelberg: Springer, 425-431.View ArticleGoogle Scholar
- Lyman JT: Complication probability as assessed from dose-volume histograms. Radiat Res Suppl. 1985, 8: S13-19.View ArticlePubMedGoogle Scholar
- Burman C, Kutcher G, Emami B, Goitein M: Fitting of normal tissue tolerance data to an analytic function. Int J Radiat Oncol Biol Phys. 1991, 21: 123-135.View ArticlePubMedGoogle Scholar
- Emami B, Lyman J, Brown A, Coia L, Goitein M, Munzenrider JE, Shank B, Solin LJ, Wesson M: Tolerance of normal tissue to therapeutic irradiation. Int J Radiat Oncol Biol Phys. 1991, 21: 109-122.View ArticlePubMedGoogle Scholar
- Niemierko A: Reporting and analyzing dose distributions: a concept of equivalent uniform dose. Med Phys. 1997, 24: 103-110. 10.1118/1.598063.View ArticlePubMedGoogle Scholar
- Meijer GJ, Van den Brink M, Hoogeman MS, Meinders J, Lebesque JV: Dose-wall histograms and normalized dose-surface histograms for the rectum: a new method to analyze the dose distribution over the rectum in conformal radiotherapy. Int J Radiat Oncol Biol Phys. 1999, 45: 1073-1080. 10.1016/S0360-3016(99)00270-9.View ArticlePubMedGoogle Scholar
- Byrne TE: A review of prostate motion with considerations for the treatment of prostate cancer. Med Dosim. 2005, 30: 155-161. 10.1016/j.meddos.2005.03.005.View ArticlePubMedGoogle Scholar
- Cheung R, Tucker SL, Lee AK, De Crevoisier R, Dong L, Kamat A, Pisters L, Kuban D: Dose–response characteristics of low- and intermediate-risk prostate cancer treated with external beam radiotherapy. Int J Radiat Oncol Biol Phys. 2005, 61: 993-1002. 10.1016/j.ijrobp.2004.07.723.View ArticlePubMedGoogle Scholar
- Ritter M: Rationale, conduct, and outcome using hypofractionated radiotherapy in prostate cancer. Semin Radiat Oncol. 2008, 18: 249-256. 10.1016/j.semradonc.2008.04.007.View ArticlePubMedPubMed CentralGoogle Scholar
- Isacsson U, Nilsson K, Asplund S, Morhed E, Montelius A, Turesson I: A method to separate the rectum from the prostate during proton beam radiotherapy of prostate cancer patients. Acta Oncol. 2010, 49: 500-505. 10.3109/02841861003745535.View ArticlePubMedGoogle Scholar
- Zavgorodni S: The impact of inter-fraction dose variations on biological equivalent dose (BED): the concept of equivalent constant dose. Phys Med Biol. 2004, 49: 5333-5345. 10.1088/0031-9155/49/23/010.View ArticlePubMedGoogle Scholar
- Bortfeld T: An analytical approximation of the Bragg curve for therapeutic proton beams. Med Phys. 1997, 24: 2024-2033. 10.1118/1.598116.View ArticlePubMedGoogle Scholar
- Russell KR, Isacsson U, Saxner M, Ahnesjo A, Montelius A, Grusell E, Dahlgren CV, Lorin S, Glimelius B: Implementation of pencil kernel and depth penetration algorithms for treatment planning of proton beams. Phys Med Biol. 2000, 45: 9-27. 10.1088/0031-9155/45/1/302.View ArticlePubMedGoogle Scholar
- Kimstrand P, Traneus E, Ahnesjo A, Grusell E, Glimelius B, Tilly N: A beam source model for scanned proton beams. Phys Med Biol. 2007, 52: 3151-3168. 10.1088/0031-9155/52/11/015.View ArticlePubMedGoogle Scholar
- Grevillot L, Bertrand D, Dessy F, Freud N, Sarrut D: A Monte Carlo pencil beam scanning model for proton treatment plan simulation using GATE/GEANT4. Phys Med Biol. 2011, 56: 5203-5219. 10.1088/0031-9155/56/16/008.View ArticlePubMedGoogle Scholar
- Jaffray DA, Lindsay PE, Brock KK, Deasy JO, Tome WA: Accurate accumulation of dose for improved understanding of radiation effects in normal tissue. Int J Radiat Oncol Biol Phys. 2010, 76: S135-139. 10.1016/j.ijrobp.2009.06.093.View ArticlePubMedPubMed CentralGoogle Scholar
- Murphy K, Van Ginneken B, Reinhardt J, Kabus S, Ding K, Deng X, Cao K, Du K, Christensen G, Garcia V, et al: Evaluation of registration methods on thoracic CT: the EMPIRE10 challenge. IEEE Trans Med Imaging. 2011, 30: 1901-1920.View ArticlePubMedGoogle Scholar
- The pre-publication history for this paper can be accessed here:http://www.biomedcentral.com/1756-6649/13/2/prepub