- Research article
- Open Access
Multiple window spatial registration error of a gamma camera: 133Ba point source as a replacement of the NEMA procedure
© Bergmann et al; licensee BioMed Central Ltd. 2008
- Received: 12 June 2008
- Accepted: 09 December 2008
- Published: 09 December 2008
The accuracy of multiple window spatial resolution characterises the performance of a gamma camera for dual isotope imaging. In the present study we investigate an alternative method to the standard NEMA procedure for measuring this performance parameter.
A long-lived 133Ba point source with gamma energies close to 67Ga and a single bore lead collimator were used to measure the multiple window spatial registration error. Calculation of the positions of the point source in the images used the NEMA algorithm. The results were validated against the values obtained by the standard NEMA procedure which uses a liquid 67Ga source with collimation.
Of the source-collimator configurations under investigation an optimum collimator geometry, consisting of a 5 mm thick lead disk with a diameter of 46 mm and a 5 mm central bore, was selected. The multiple window spatial registration errors obtained by the 133Ba method showed excellent reproducibility (standard deviation < 0.07 mm). The values were compared with the results from the NEMA procedure obtained at the same locations and showed small differences with a correlation coefficient of 0.51 (p < 0.05). In addition, the 133Ba point source method proved to be much easier to use. A Bland-Altman analysis showed that the 133Ba and the 67Ga Method can be used interchangeably.
The 133Ba point source method measures the multiple window spatial registration error with essentially the same accuracy as the NEMA-recommended procedure, but is easier and safer to use and has the potential to replace the current standard procedure.
- Gamma Camera
- Lead Collimator
- 67Ga Source
- Dual Isotope Imaging
- 133Ba Source
The ability of a gamma camera to perform quantitative dual isotope imaging depends critically on its ability to accurately position photons of different energies when imaged through different photo peak energy windows. This is particularly true when localizing small organs by subtraction scintigraphy, such as the parathyroid glands[1, 2] or lung perfusion-ventilation imaging [3, 4]. Poor multiple window spatial resolution (MWSR) also deteriorates spatial resolution when imaging a single radionuclide that has multiple photo peaks (e.g. 67Ga, 201Tl, 111In). Both the NEMA  and the IEC  standards use the same procedure to assess the accuracy of MWSR. It consists in positioning a collimated 67Ga source on the detector surface of the gamma camera and determining differences in the positions of the source image at different energy window settings. The procedure has two main shortcomings, one is that the collimator used is heavy (ca. 1.2 kg) and therefore potentially dangerous to the fragile detector surface. The other one is the use of a liquid 67Ga source which often has to be purchased specifically for this test. We developed an alternative method to assess the accuracy of multiple window spatial registration using a long-lived 133Ba point source collimated by a small lead collimator.
The experiments were carried out on three camera heads of different crystal thicknesses (9.5, 15.9, and 25.4 mm) with large rectangular field of views.
For each position and collimator geometry two images for the low energy photons and the high energy photons of either the 67Ga or 133Ba source were acquired for 120 s in a 512 × 512 matrix, pixel size 1.1 mm. The peak energy settings were 93 and 300 keV for 67Ga and 80 and 356 keV for 133Ba, respectively, with all energy windows set to 20%. The position of the source in an image was determined according to the NEMA algorithm by calculating the centre of gravity of the intensity within a square region with a side length of four times the full width at half maximum (FWHM) of the peak. To avoid potential errors from differences in region positioning a program written in Matlab (Version 2006a, The Mathworks Inc., Natick, MA) was developed in which the approximate position of the point source was first identified manually, followed by a fully automatic placement of the region of interest and calculations of the registration offsets. In order to identify optimum collimator dimensions, initial experiments were made using collimator bore diameters of 2, 3, 4 and 5 mm as well as the 133Ba source without collimation. Since the collimators with 2 and 3 mm bores were inferior to the collimators with larger holes with respect to sensitivity (see results section), the localization comparisons with the 67Ga values were then made with the collimators with a hole size of 4 and 5 mm and with the 133Ba source without collimation.
In order to facilitate the comparison of the results, the differences between low and high energy positions for the two radionuclides were displayed visually. The similarity between the locations of the two radionuclides were analyzed further by performing regression analyses using the statistical software SPSS (Version 14.0.1, SPSS Inc., Chicago, IL). Statistical significance was assumed for a p value < 0.05. A Bland-Altman analysis  was performed for statistically significant regressions.
Reproducibility was tested both for the 133Ba method using the 5 mm collimator and the 67Ga method by acquiring low and high energy images in one template position. For each configuration 10 measurements were carried out. Between each measurement the source with collimator was completely removed and positioned anew. Reproducibility and accuracy of the centre of the 133Ba source were tested by acquiring low and high energy images in one template position, for the 5 mm collimator. Eight acquisitions were performed. Between each of these acquisitions the point source was rotated by 45°.
Reproducibility expressed as standard deviation of the 10 measurements was 0.07 mm for the 133Ba and 0.17 mm for the 67Ga displacement. When rotating the 133Ba source in the collimator, reproducibility expressed as standard deviation of the 8 measurements was 0.2 mm. This was due to a small displacement of the point source activity by 0.1 mm from the geometric midpoint of the circular disk containing the source. The additional error was avoided in the subsequent measurements by using always the same angular orientation of the disk with respect to the template.
NEMA multiple window spatial registration (MWSR) error for the 3 camera heads
MWSR Error (mm)
Detector Thickness (mm)
Modern gamma cameras employ several corrections to improve non-linearity and non-uniformity. The correction parameters are obtained by tuning, carried out for the different radionuclide energies separately . The basic performance parameters of a gamma camera are primarily measured with 99mTc. Good accuracy of multiple window spatial resolution ensures that performance parameters such as spatial linearity or pixel calibration are not dependent on the energy (i.e. magnitude of light output). The spatial registration error of modern gamma cameras is usually specified as being below 1 mm. This limit of accuracy is acceptable for all applications of dual isotope imaging or single isotope imaging with multiple energy windows, and is achieved after tuning of the gamma camera for the particular energies in question. The proposed method for assessing the multiple windows spatial registration error has shown to give similar results as the NEMA method, but the measurement procedure is much easier and, because of the light weight collimator used with the 133Ba source, potentially safer to carry out. The different collimators tested show that holes of 4 and 5 mm diameter provide essentially the same results, although the better counting efficiency of the latter makes it the better choice. The Bland-Altman analysis shows that the limits of agreement between the 133Ba and the 67Ga Method are approximately one third of the usual limit for the spatial registration error of a modern gamma camera. This indicates that both methods can be used interchangeably. Although even the uncollimated 133Ba source gives acceptable results, the high fraction of counts outside the region of interest due to a greater divergence of the photons incident on the crystal renders the analysis more complicated since it is not possible to sum all individual point source images for joint analysis because of overlapping intensities as can be done with the collimated images. Furthermore, the source strength of the 133Ba source is more critical because the uncollimated count rate may easily exceed the recommended maximum count rate of 30 kcps for the measurement of performance parameters. A drawback of the method may be seen in the fact that 133Ba has only 2 photon energies, whereas 67Ga offers 3 including one close to the energy of 99mTc. However, we believe it to be sufficient to measure at a low and a high energy covering the range of all commonly used radionuclide energies. This is consistent with the NEMA standard in which the low and the high energy of 67Ga are to be used if only 2 energy windows are available. Another advantage of the proposed method is that due to the long half-life of 133Ba a standard acquisition procedure including a fixed optimal acquisition time can be used which makes it easy to carry out the test routinely, such as after recalibration of the gamma camera.
Multiple window spatial registration can alternatively be measured with a 133Ba point source and a mini-collimator weighing less than 8 percent than the standard NEMA collimator. The spatial displacement errors are quantitatively the same as those obtained using the standard NEMA procedure employing a liquid 67Ga source and a heavy collimator. Advantages of the alternative method are the use of a long-lived radioactive nuclide, no need for handling a liquid radioactive source, and the use of a light-weight collimator which makes the procedure much easier and potentially safer to carry out than the standard NEMA procedure. Possible disadvantages are the lack of an energy peak close to 140 keV and the need to buy and store a long lived 133Ba source. The 133Ba point source method has the potential to replace the NEMA procedure for the measurement of multiple window spatial registration errors.
- Staudenherz A, Abela C, Niederle B, Steiner E, Helbich T, Puig S, Kaserer K, Becherer A, Leitha T, Kletter K: Comparison and histopathological correlation of three parathyroid imaging methods in a population with a high prevalence of concomitant thyroid diseases. Eur J Nucl Med. 1997, 24: 143-149. 10.1007/BF02439546.View ArticlePubMedGoogle Scholar
- Wakamatsu H, Noguchi S, Yamashita H, Yamashita H, Tamura S, Jinnouchi S, Nagamachi S, Futami S: Parathyroid scintigraphy with 99mTc-MIBI and 123I subtraction: a comparison with magnetic resonance imaging and ultrasonography. Nucl Med Commun. 2003, 24: 755-762. 10.1097/00006231-200307000-00004.View ArticlePubMedGoogle Scholar
- Kohn H, Mostbeck A, Bachmayr S, Eber O, Galvan G, Holm C, Konig B, Lind P, Markt B, Ogris E: 99mTc-DTPA aerosol for same-day post-perfusion ventilation imaging: results of a multicentre study. Eur J Nucl Med. 1993, 20: 4-9. 10.1007/BF02261238.View ArticlePubMedGoogle Scholar
- Magnant J, Vecellio L, de Monte M, Grimbert D, Valat C, Boissinot E, Guilloteau D, Lemarie E, Diot P: Comparative analysis of different scintigraphic approaches to assess pulmonary ventilation. J Aerosol Med. 2006, 19: 148-159. 10.1089/jam.2006.19.148.View ArticlePubMedGoogle Scholar
- National Electrical Manufacturers Association: NEMA Standards Publication NU 1-2007: Performance measurements of gamma cameras. Rosslyn, VA. 2007Google Scholar
- IEC: IEC 60789: Medical electrical equipment – Characteristics and test conditions of radionuclide imaging devices – Anger type gamma cameras. Geneva. 2006Google Scholar
- Bland JM, Altman DG: Statistical methods for assessing agreement between two methods of clinical measurement. Lancet. 1986, 1: 307-310.View ArticlePubMedGoogle Scholar
- Cherry S, Sorenson J, Phelps M: Physics in Nuclear Medicine. 2003, Elsevier Science (USA), 235-239.Google Scholar
- The pre-publication history for this paper can be accessed here:http://www.biomedcentral.com/1756-6649/8/6/prepub
This article is published under license to BioMed Central Ltd. This is an Open Access article distributed under the terms of the Creative Commons Attribution License (http://creativecommons.org/licenses/by/2.0), which permits unrestricted use, distribution, and reproduction in any medium, provided the original work is properly cited.